The Potentiometer Pop End Fitting used with Aim loggers measures displacement between two points using a sensor (linear potentiometer) directly connected to the measure points.
This potentiometer can measure linear displacements like:
- dampers compression or extension
- steering rotation measured through the rack displacement
When installing the sensor:
- be very careful avoiding possible bending of the internal cylinder; these bendings, occurring when over tightening the screws or in case of incorrect mounting can seriously damage the sensor •
- extract the internal cylinder for about 5 mm (0.2 inches) from the sensor lower boundary position.
Please note: do not use this sensor to measure distances beyond the potentiometer maximum travel.
The car/bike linear potentiometer can be connected to any analogue channel of AiM loggers.
Software setup – Suspensions
To load the potentiometer in the logger configuration run the software and select the configuration you are going to load it on.
Enter the configuration (in the example MXL2 03) and the related "Channels" layer.
- Select the channel where to set the potentiometer on – in the example channel 6 (1) and fill in the panel that shows up
- Function: "Position" (2)
- Sensor: "Position Pot. AutoCal" (3 – this implies that the potentiometer will be auto-calibrated as shown in the following pages)
- Fill in the other fields
- Fill "Total Potentiometer travel" box with the potentiometer travel in mm – in the example we used a 50mm travel potentiometer (4)
- Click "Save"
When the software comes back to "Channels" layer the potentiometer has been set on the desired channel as shown here below.
• Transmit the configuration to the logger pressing "Transmit" on the top keyboard
To auto-calibrate the potentiometer:
- enter "All" layer and press "Device" (1)
- select the logger – in the example MXL2 ID 410 (2)
- in "Live Measures" layer, keeping the potentiometer in its zero position, select the channel where the potentiometer has been set – in the example channel 6 (3)
- press "Auto Calibrate" (4)
- Keep the potentiometer in its zero position as shown here below
- Press "Auto calibrate All"